Accuracy class 

0.1  
Torque measuring system (rotating)  
Nominal (rated) torque Mnom 
kNm 
100 
125 
150 
200 
250 
300 
Nominal (rated) rotational speed 
rpm 
3000 
2000  
Linearity deviation including hysteresis, related to nominal (rated) sensitivity Frequency output For a max. torque in the range: 

 
between 0% of Mnom and 20% of Mnom 
% 
≤±0.03  
> 20% of Mnom and 60% of Mnom 
% 
≤±0.065  
> 60% of Mnom and 100% of Mnom 
% 
≤±0.1  
Voltage output For a max. torque in the range: 

 
between 0% of Mnom and 20% of Mnom 
% 
≤±0.03  
> 20% of Mnom and 60% of Mnom 
% 
≤±0.065  
> 60% of Mnom and 100% of Mnom 
% 
≤±0.1  
Rel. standard deviation of repeatability, per DIN 1319, related to the variation of the output signal 

 
Frequency output 
% 
≤±0.02  
Voltage output 
% 
≤± 0.02  
Temperature effect per 10 K in the nominal (rated) tempera ture range on the output signal, related to the actual value of the signal span 

 
Frequency output 
% 
≤±0.1  
Voltage output 
% 
≤±0.1  
on the zero signal, related to the nominal sensitivity 

 
Frequency output 
% 
≤±0.07  
Voltage output 
% 
≤±0.07  
Nominal sensitivity (spread between torque = zero and nominal torque) 

 
Frequency output 10 kHz / 60 kHz / 240 kHz 
kHz 
5/30/120  
Voltage output 
V 
10  
Sensitivity tolerance (deviation of the actual output quantity at Mnom from the nominal (rated) sensitivity) 

 
Frequency output 
% 
±0.1  
Voltage output 
% 
±0.1  
Output signal at torque = zero 

 
Frequency output 
kHz 
10/60/240  
Voltage output 
V 
0  
Nominal (rated) output signal 

 
Frequency output 

 
at positive nominal (rated) torque 
kHz 
15 1) / 90 2) / 360 3) (5 V balanced 4))  
at negative nominal (rated) torque 
kHz 
5 1)/ 30 / 120 3) (5 V balanced 4))  
Voltage output 

 
at positive nominal (rated) torque 
V 
+10  
at negative nominal (rated) torque 
V 
10  
Load resistance 

 
Frequency output 
kΩ 
≥2  
Voltage output 
kΩ 
≥10 